In closed tanks, the DP transmitter must compensate for the pressure of the vessel above the liquid surface. So the LP side of transmitter is connected to the tank. Here in this post we are discussing about calibration of DP transmitter for a closed tank with dry leg & constant level application for density measurement. This calculation tool only used for constant level application.

Density Measurement using DP Transmitter for closed tank with dry leg 
The minimum and maximum process specific gravities can be obtained from process datasheets.
The formulas for calculating the tranmitter LRV and URV are;
HP Side/LRV/4ma = LSmin
LP Side/URV/20ma = LSmax
Where
L = Closed tank maximum measured level
Smax = Maximum process specific gravity
Smin = Minimum process specific gravity
Note: This calculation is only used for closed tank with dry leg and for constant level application.
Similar Articles:
 Density Measurement using DP Transmitter for closed tank with wet leg
 Density Measurement for open tank with constant level application
 Closed Tank DP Type Level Measurement with Dry leg and transmitter installed below tapping point
 Closed Tank DP Type Level Measurement with wet leg and transmitter installed below tapping point
 DP Type Level Measurement at transmitter installed below tapping point for open tank
 DP Type Level Measurement at transmitter installed above tapping point for open tank
 Interface Level Measurement using DP Transmitter
 Closed Tank Remote Seal Capillary type DP Transmitter
 Zero Suppression and Zero Elevation Calculations
 Open Tank DP Level Transmitter Calculations