Home » Calculate the 0%, 50%, and 100% Calibration Points
Practical Questions

# Calculate the 0%, 50%, and 100% Calibration Points

Calculate the 0%, 50%, and 100% calibration points for the DP transmitter to measure liquid level in this vessel, given a range of 0 to 30 feet and a process specific gravity of 0.85.

Note the “wet” leg filled with fluid different than the process (SG = 1.1), 35 feet tall:

In this example, the transmitter output loop current range is 10 to 50 mA.

## DP transmitter Calibration Points Calculate the equivalent ΔP range (inch H2O) and loop current (mA) for 0%, 50%, and 100% of process variable.

ΔP range (inch H2O)

0% = -462

50% = -309

100% = -156

Loop Current (mA)

0% = 10

50% = 30

100% = 50

Credits: Tony R. Kuphaldt

#### Practical Process Control System Questions & Answers – 6

This website uses cookies to improve your experience. We'll assume you're ok with this, but you can opt-out if you wish. Accept Read More

WordPress Image Lightbox
Send this to a friend