# Density Measurement using DP Transmitter for closed tank with wet leg

In closed tanks, the DP transmitter must compensate for the pressure of the vessel above the liquid surface. So the LP side of transmitter is connected to the tank. Here in this post we are discussing about calibration of DP transmitter for a closed tank with wet leg & constant level application for density measurement. This is only used for constant level application.

### Density Measurement using DP Transmitter

 Density Measurement using DP Transmitter for closed tank with wet leg

The minimum and maximum process specific gravities can be obtained from process datasheets.

The formulas for calculating the tranmitter LRV and URV are;

HP Side/LRV/4ma = L.Smin – H.Sref

LP Side/URV/20ma = L.Smax – H.Sref

Where

L = Closed tank maximum measured level
H = Height of wet leg
Smax = Maximum process specific gravity
Smin = Minimum process specific gravity
Sref = wet leg or reference leg specific gravity

Note: This calculation is only used for closed tank with wet leg and for constant level application.

Be the first to get exclusive content straight to your email.
We promise not to spam you. You can unsubscribe at any time.

### 4 thoughts on “Density Measurement using DP Transmitter for closed tank with wet leg”

1. Need to learn Instrument .