Instrumentation engineering root cause analysis of re-heating furnace too low-temperature readings after a failed thermocouple change.
|Article Type:||Root Cause Analysis (RCA)|
|Author:||S. Raghava Chari|
Note: This root cause analysis (RCA) is from real-time scenarios that happened in industries during the tenure of two or three decades ago. These articles will help you to improve your troubleshooting skills and knowledge.
Too Low-Temperature Readings
The re-heating furnace supervisor complained about too low-temperature readings after a failed thermocouple (TC) change.
The instrument crew repeatedly found correct recorder calibration. Hence, the furnace supervisor asked the author to resolve to enable using the furnace to increase rolling.
The author’s Optical Pyrometer check confirmed furnace temperature around 150O C more than the temperature recorder reading.
He complimented the furnace supervisor for his major disasters avert judgment – slabs if any charged into the furnace would have melted and made it unusable.
Cleaning the mess and repairing the furnace would have taken several weeks and reduced rolling mill output 10 to 15%.
He sketched figs A and B and explained to the crew how the error happened. He instructed them to take care of the following when making TCs and installing them in the field:
- Connect the thermocouple wires to head (Hd) terminals making sure of polarities
- Connect extension leads at thermocouple (TC) head, intermediate JBs if any and at instrument terminals making sure of polarities. In short, correct polarities are essential all the way.
With the Thermocouple (TC) and Extension
Wires correct polarities all the way the recorder read correct and rolling resumed.
Author: S. Raghava Chari
Do you face any similar issues? Share with us through the below comments section.
If you liked this article, then please subscribe to our YouTube Channel for Instrumentation, Electrical, PLC, and SCADA video tutorials.
- Target Flow Transmitters Failed
- Multistage Flash Drums Readings
- Rotameter Caused Pipe Weld Cracks
- Level Instruments Stopped Working
- Differential Pressure Transmitter Problem