A flow switch is one detecting the flow of some fluid through a pipe. Flow switches often use “paddles” as the flow-sensing element, the motion of which actuates one or more switch contacts.
The “normal” status of a switch is the resting condition of no stimulation. A flow switch will be in its “normal” status when it senses minimum flow (i.e. no fluid moving through the pipe). For a flow switch, “normal” status is any fluid flow rate below the trip threshold of the switch.
A simple paddle placed in the midst of a fluid stream generates a mechanical force which may be used to actuate a switch mechanism, as shown in the following photograph:
Like all other process switches, flow switches exhibit deadband (also called differential ) in their switching action. A flow switch that trips at 15 GPM rising, for example, will not re-set at 15 GPM falling. That switch would more likely reset at some lower flow rate such as 14 GPM. With the “trip” and “reset” points being different, the switch avoids unnecessary cycling if the process flow rate hovers near one critical value.
- Temperature Switch Principle
- Basics of Pressure switches
- Flow Nozzle Principle
- Normally-open and Normally-closed Switch Contacts
- Working Principle of Flow Switch with Animation
- Define Normal Status of a Process Switch ?
- Pressure Switch Calibration Procedure
- How to Test a Pressure Switch ?
- Float level switch Principle
- What is a Velocity-based Flowmeter ?