#### Explanation of the terms: Transmitter’s Turn down (TD) ratio, set span and zero based span

Consider a Pressure Transmitter and we will discuss some important parameters related to it.

**Range : **is the measurement limit and covers from the minimum to the maximum pressure that the transmitter can measure, e.g., 0 to 5080 mmH2O. The maximum span is 5080 mmH2O.

** URL (Upper Range Limit) : **is the highest pressure at which the transmitter was designed to measure, respected the sensor upper range limit.

** LRL (Lower Range Limit) : **is the lowest pressure at which the transmitter was designed to measure, respected the sensor lower range limit.

** URV (Upper Range Value) : **is the maximum pressure at which the transmitter calibrated.

** LRV (Lower Range Value) : **is the minimum pressure at which the transmitter calibrated.

** Span (Range Calibrated) **: the work range where the calibration is done is known as span, for example, from 500 to 3000 mmH2O, where the span is 3000-500 = 2500 mmH2O. The Span is equal to URV – LRV.

** Turndown(TD) or Rangeability : **is the relation between the maximum pressure (URL) and the minimum measured pressure (minimum calibrated span). For example, a transmitter range is 0-5080 mmH2O and will be used on 10:1, indicating which transmitter will measure 0 to 508 mmH2O. TD = URL/ Calibrated Span.

**Case 1: **

**|Lower range value| ≤ |Upper range value|**

Figure : 1000 mbar (14.5 psi) sensor

Here,

1 : Set span

2 : Zero based span

3 : Nominal value ≅ Upper range limit (URL)

4 : Nominal measuring range

5 : Sensor measuring range

LRL : Lower range limit

URL : Upper range limit

LRV : Lower range value

URV : Upper range value

**Example:**

Lower range value (LRV) = 0 mbar

Upper range value (URV) = 100 mbar (1.5 psi)

Nominal value (URL) = 1000 mbar (14.5038 psi)

**Turn down:**

TD = URL /|URV| = 10 : 1

**set span:**

URV – LRV = 100 mbar (1.5 psi)

This span is based on the zero point.

**Case 2: **

**|Lower range value| ≥ |Upper range value|**

Figure : 1000 mbar (14.5 psi) sensor

**Example:**

Lower range value (LRV) = -300 mbar (-4.5 psi)

Upper range value (URV) = 0 mbar (0 psi)

Nominal value (URL) = 1000 mbar (14.5038 psi)

**Turn down:**

TD = URL /|URV| = 3.33 : 1

**set span:**

URV – LRV = 100 mbar (1.5 psi)

This span is based on the zero point.

**Also Read : Importance of Turndown Ratio**

Reference : endress hauser

#### Similar Articles:

- Instrumentation Engineer’s Calibration Mistakes
- DP Level Transmitter Elevation and Suppression Corrections
- Smart Transmitter Calibration Tutorial Part 2
- Instrument Turndown Ratio
- Important Terms in Calibration
- Interface Level Measurement using DP Transmitter
- Zero Suppression and Zero Elevation Calculations
- Input-Output Relationships of Transmitters
- Remote Seal Transmitters Ranging Calculation
- Closed Tank Level Measurement using DP Transmitters

## 3 comments

Hi All,

In the above, please elaborate the effect of Turn Down Ratio of higher or lower during operation or selection of transmitters.

Hi, Please search the website with word “Turndown” its already there. Thanks

Hi , I have certain doubts in case 2

1. If TD = URL/URV , then it should be 1000/0 as per the formula mentioned

2. Span = URV – LRV, then it should be 0-(-300), which is 300.

Please correct me if am wrong