Flow Meter Turndown Ratio
When specifying a flowmeter, accuracy is a necessary requirement, but it is also essential to select a flowmeter with sufficient range for the application.
‘Turndown’ or ‘turndown ratio’, ‘effective range’ or ‘rangeability’ are all terms used to describe the range of flowrates over which the flowmeter will work within the accuracy and repeatability of the tolerances.
Also Read : Importance of Flow Meter Accuracy
A particular steam system has a demand pattern as shown in below Figure. The flowmeter has been sized to meet the maximum expected flowrate of 1 000 kg/h.
The turndown of the flowmeter selected is given as 4:1. i.e. The claimed accuracy of the flowmeter can be met at a minimum flowrate of 1000 ÷ 4 = 250 kg/h.
When the steam flowrate is lower than this, the flowmeter cannot meet its specification, so large flow errors occur. At best, the recorded flows below 250 kg/h are inaccurate – at worst they are not recorded at all, and are ‘lost’.
In the example shown in Figure, ‘lost flow’ is shown to amount to more than 700 kg of steam over an 8 hour period. The total amount of steam used during this time is approximately 2700 kg, so the ‘lost’ amount represents an additional 30% of total steam use. Had the steam flowmeter been specified with an appropriate turndown capability, the steam flow to the process could have been more accurately measured and costed.
If steam flow is to be accurately metered, the user must make every effort to build up a true and complete assessment of demand, and then specify a flowmeter with:
- The capacity to meet maximum demand.
- A turndown sufficiently large to encompass all anticipated flow variations.
Source : spiraxsarco
- Impact of Flow Meter Accuracy
- Instrument Turndown Ratio
- Orifice Plate Turndown ratio
- What is a Differential Pressure Flow meter?
- Pitot Tube Working Principle
- Flow Transmitter Rangeability
- Spring Loaded Variable Area Flowmeter Working Principle
- Swirl Flow Meters Working Principle
- Transmitter Turndown Ratio, Set Span and Zero Span
- How to Select a Flow Meter