**Hydrostatic Level Measurement**

This method rely on the pressure of the measured liquid head to provide level indication. The pressure exerted by a column of liquid is equal to the height of the column, h, times the specific gravity of the liquid, sg. (P=h⋅(sg), where sg=ρ⋅g). Thus, from this we can calculate the level of the liquid in a vessel or tank, (i.e. the h), if we know the pressure at the bottom of the tank and the specific gravity of the liquid, sg.

By referring to the figure above, the weight of one cubic foot container of water is seen to be 62.4 pounds (from its weight), and this force is exerted over the surface of the bottom of the container.

Area, A = 12 in × 12 in = 144 in² (Since 1 ft = 12 in)

Pressure exerted on the area, P = F/A = 62.4 lbs/144 in² = 0.433 lbs/in²

We calculate equivalent level value based on the measured pressure and pressure directly proportional to the level. if pressure increases that means level in the column/tank increase and vice-versa.

**Hydrostatic Level Measurement Animation**

#### Similar Articles:

- Basics of Hydrostatic Level Measurement
- Open Tank Level Measurement using DP Transmitters Animation
- Hydrostatic Level Measurement – Temperature related Density Change Effects
- Effect of Temperature & Pressure on Level Measurement
- Differential Pressure Level Transmitter Errors
- Level and Density Measurement Objective Questions
- Basics of Pressure Measurement
- Closed Tank DP Type Level Measurement with wet leg and transmitter installed below tapping point
- Liquid Level Measurement using Pressure Transmitter Questions
- Zero Suppression and Zero Elevation Calculations

## 1 comment

Hello Sir,

I am very interested to read instrumentation article. In this website have very useful information. Thanks for make like this very useful information to you. I would like to ask download link for animation, picture and article. If it is there means, its very useful if offline.