﻿ Determine the Instrument Calibration Error
Home » Determine the Instrument Calibration Error
Practical Questions

# Determine the Instrument Calibration Error

Suppose an electronic pressure transmitter has an input range of 0 to 100 PSI and an output range of 4 to 20 mA. When subjected to a 5-step up-and-down “As-Found” calibration test, it responds as such:

Sketch this instrument’s ideal transfer function on the graph below, along with its actual transfer function graph based on the measured values recorded above. Then, determine what kind of calibration error it has (zero shift, span shift, hysteresis, and/or linearity):

Finally, identify how this calibration error might be corrected. What steps or procedures would you follow to rectify this problem?

How might the other three calibration errors appear when graphed?

What purpose is served by doing an up-and-down test? Why not just check the instrument’s response in one direction only?

Which constant in the y = mx + b linear equation represents zero, and which represents span?

This instrument has a zero shift error, but not a span shift or linearity error.

#### A linearity error would look something like this (not a straight line):

Span errors, by contrast, usually require multiple adjustments of the “zero” and “span” screws while alternately applying 0% and 100% input range values to check for correspondence at both ends of the linear function.

Credits: Tony R. Kuphaldt

#### Programmable Logic Controller (PLC) Questions and Answers – 1

This website uses cookies to improve your experience. We'll assume you're ok with this, but you can opt-out if you wish. Accept Read More

WordPress Image Lightbox
Send this to a friend