Velocity Flow meter
The Law of Continuity for fluids states that the product of mass density (ρ), cross-sectional pipe area (A) and average velocity (v) must remain constant through any continuous length of pipe:
If the density of the fluid is not subject to change as it travels through the pipe (a very good assumption for liquids), we may simplify the Law of Continuity by eliminating the density terms from the equation:
The product of cross-sectional pipe area and average fluid velocity is the volumetric flow rate of the fluid through the pipe (Q = Av). This tells us that fluid velocity will be directly proportional to volumetric flow rate given a known cross-sectional area and a constant density for the fluid flowstream. Any device able to directly measure fluid velocity is therefore capable of inferring volumetric flow rate of fluid in a pipe. This is the basis for velocity-based flowmeter designs.
- Why Flow Directly Proportional to Square Root of ΔP ?
- Pressure based Flowmeters
- Coriolis flowmeter Advantages and limitations
- What is a Vortex flowmeter ?
- Annubar Flow Meter Working Principle
- V Cone Flow Meter Working Principle
- Basics of Control Valve Sizing
- What is a Turbine flow meter ?
- Basics of Flow measurement
- Difference between Different Types of Flow Meters