Pressure Gauge Functional Testing
Test Equipment Required
- Dead Weight Tester
- Test Gauge
Also Read : What is a Dead Weight Tester ?
Typical Test Set-up
a. Check and confirm pressure gauges range is as per data sheets.
b. Hook-up instruments as per typical test set-up.
c. With calibration pressure at zero, confirm the gauge indicates zero.
d. Apply calibration pressure at 100% of range. confirm the gauge indicates maximum span.
e. Carry out 5 points calibration check at 0%, 25%, 50%, 75%, 100% of span for rising and falling.
f. Record all results in Instrument Calibration (Indicator, Recorder) Checksheet.
Note: Hydraulic pump/ test gauge to be used in lieu of deadweight tester for site Calibration.
- Sand Probe Functional Testing
- Differential Pressure Switch Calibration Procedure
- Self Actuated Pressure Control Valve Functional Testing
- Solenoid Valve Functional Testing
- Pig Signalling Switches Functional Testing
- What is a Dead Weight Tester ?
- Pressure Safety Valves Functional Testing
- Instrument Calibration Multiple Choice Questions
- Pressure Switch Calibration Procedure
- Motor Operated Valve Functional Testing