In any process industry, level measurement is very important for safety and process purpose.

Level can be measured by two methods.

**Direct method****Indirect method**

Here we discuss about indirect method and which is using DP transmitter for level measurement.

A Differential pressure transmitter is a common and well understood technology for liquid level measurement. If the tank is closed or pressurized, a DP measurement must be made to compensate for the vessel pressure.

** DP Transmitter Level Measurement for Open Tank**

Suppose there is a open tank in which a DP transmitter installed for the measurement of level then HP leg of the DP transmitter is connected with bottom of tank and LP leg is open in atmosphere. Assume process fluid specific gravity in the tank is 1.0

Definition of specific gravity: Specific gravity is a measure of density relative to the density of a reference substance. Reference substance is taken water.

For DP Transmitter Configuration, we have to find out Zero Level & Span Level. Accordingly we have to configure Lower Range Value (LRV) and Upper Range Value (URV) using HART communicator.

**DP Transmitter installed at the exact HP tapping point**

**At Zero level** = 0 mmwc

**At Span level** = H x Specific gravity

= 500 x 1.0

= 500 mmwc

**Then Range** = 500 – 0 = 500 mmwc

So, we have to set Lower Range Value (LRV) = 0 mmwc and Upper Range Value (URV) = 500 mmwc in the DP Transmitter using HART communicator.

**DP Transmitter Installed Below Tapping Point for Open Tank**

If transmitter is installed below the tank then it’s called zero suppression then calculation is below:

When zero suppression is used then H1 height always filled with process fluid.

**At zero level (LRV) = H1 x specific gravity**

= 100 x 1.0

= 100 mmwc

**At 100% level (URV) = (H1 + H2) x specific gravity**

= (100 + 500) x 1.0

= 600 mmwc

**Range = URV – LRV** = 600 – 100 = 500 mmwc

So, we have to set Lower Range Value (LRV) = 100 mmwc and Upper Range Value (URV) = 600 mmwc in the DP Transmitter using HART communicator.

##### Article by :

Ashish Agrawal

#### Similar Articles:

- DP Transmitter Dry Leg & Wet Leg Calculations
- Open Tank Interface Level Measurement
- Open Tank Level Measurement using DP Transmitter
- Calibrating Level Transmitter with Remote Seals
- DP Type Level Measurement at transmitter installed below tapping point for open tank
- DP Type Level Measurement at transmitter installed above tapping point for open tank
- Closed Tank Level Measurement using DP Transmitters
- Closed Tank DP Type Level Measurement with wet leg and transmitter installed below tapping point
- Density Measurement for open tank with constant level application
- Closed Tank DP Type Level Measurement with Dry leg and transmitter installed below tapping point

## 1 comment

Very good knowledge…..