A Pitot tube measuring the flow of glycerin (78.6 lb/ft3 ) produces a differential pressure of 35 inches W.C. at a mass flow rate of 450 pounds per minute.
Calculate the differential pressure produced by this same Pitot tube at a flow rate of 310 lb/min.
Calculate the mass flow rate at a differential pressure of 12.6 ”WC.
If the glycerin density happens to change from 78.6 lb/ft3 to 75.1 lb/ft3 , calculate the mass flow rate at a differential pressure of 24.8 ”WC.
Differential pressure produced by this same Pitot tube at a flow rate of 310 lb/min = 16.61 ”WC
Mass flow rate at a differential pressure of 12.6 ”WC = 270 lb/min
Mass flow rate at DP of 24.8 ”WC, assuming density of 75.1 lb/ft3 = 370.3 lb/min
More DP Transmitter Questions
- Smart DP Transmitter Flow Meter Calculations
- DP Transmitter Error Calculations
- Flow Transmitter Readings mismatch with DCS
- DP Transmitter Calibration Errors
- Mass Flow Calculations
- Volumetric Flow Calculations
- Pitot Tube Working Principle
- Annubar Flow Meter Working Principle
- Instrument Calibration Multiple Choice Questions
- Orifice Plate Turndown ratio