One of the simplest classes of temperature sensor is one where temperature effects a change in electrical resistance. With this type of primary sensing element, a simple ohmmeter is able to function as a thermometer, interpreting the resistance as a temperature measurement:
Thermistors are devices made of metal oxide which either increase in resistance with increasing temperature (a positive temperature coefficient) or decrease in resistance with increasing temperature (a negative temperature coefficient).
RTDs are devices made of pure metal wire (usually platinum or copper) which always increase in resistance with increasing temperature. The major difference between thermistors and RTDs is linearity: thermistors are highly sensitive and nonlinear, whereas RTDs are relatively insensitive but very linear. For this reason, thermistors are typically used where high accuracy is unimportant. Many consumer-grade devices use thermistors for temperature sensors.
Credits : Tony R. Kuphaldt – Creative Commons Attribution 4.0 License
- Temperature Coefficient of Resistance (α)
- Difference Between RTD, Thermocouple and Thermistor
- Resistance Temperature Detectors (RTD) Construction
- Thermistor Working Principle
- Calculate Temperature Coefficient of RTD’s
- What is RTD Sensitivity ?
- RTD Standards
- Thermal Conductivity Analyzers Questions & Answers
- RTD Wheatstone Bridge Circuits
- RTD Calculator : Calculate temperature from resistance